Yes, but only in the start up or after changing the configuration.Yes, this can only occur in the first log block created in the start up of the logger or after making a change in the configuration, the rest of the blocks will always contain the same amount of data.
Yes.Yes, it is compatible with SD or SDHC cards, but they must be formatted in FAT or FAT32. It is not compatible with the NTFS format.
The card is not inserted or is broken.This error indicates that the SD card has not been detected, either because it has not been inserted or because it is broken.
Check that there is a card inserted, if so, check its operation inserting it in another device.
The card does not allow recording data.The card inserted does not allow recording data, either because it is broken or because it is full.
Try to empty the data or check its operation inserting it in another device.
Format error in the inserted card.The inserted card does not have the correct format. Check that it is formatted in FAT or FAT32.
Card write-protectedThe card inserted has the protection against writing activated, deactivate it and try again.
Configuration protected by password.A password has been inserted to protect the configuration of the device (Parameter L5).
Insert the correct code to access (From 0 to 99). If you do not know the code, consult your installer.
If the r6 parameter (automatic time change) is configured as "Yes", it is not necessary.If the r6 parameter (automatic time change) is configured as "Yes", it is not necessary to carry out any action, the time will be automatically updated.
If it is configured as "No", you must change the time manually, accessing the r1 and r2 parameters.
For more information, consult the device's instructions.
Indicates an error in probe 1 (or 2).The probe 1 (or 2) is not correctly connected, or has broken, or is out of range. Check that the probe used covers the adequate temperature range. Check the connection of the probe or replace it if necessary.
The input shown is disabled.The input shown is disabled.
If this input does not have any probe connected, you can prevent it from being displayed with parameter P8.
The top / bottom log of the block has been reached.The top / bottom log of the block has been reached in the display menu of the data log. Press the up or down button to go to the next log.
The device is reading its internal memory.The device is reading its internal memory, this process can last longer depending on the amount of the contained data.
The internal memory of the logger is empty.The internal memory of the logger is empty, no data has been logged yet.
Can I configure the device so that it generates an alarm every time the chamber door stays open for longer than 5 minutes?
One of the digital inputs should be configured as digital input.Yes, it requires connecting the door microswitch of the chamber to one of the inputs and configure it as a digital input.
If you have connected the microswitch input 2, you must configure the device as shown:
-Parameter i11. If the door micro closes its contacts on opening the door, it must be configured in 3, if on the contrary it opens its contacts on opening the door, it must be configured in 4.
-Parameter A13=5 (min.)
-Parameter A14. If you want the alarm to activate an acoustic signal, you must configure in 1, if you want to activate the alarm relay, you must configure in 2 and if you want it to do both things (signal + relay) you must configure in 3.